Software version 1.11 will include all enhancements of previous releases in addition to the new features listed below. This version will not in any way affect the performance of your Neon™ Transfection System except for the specific changes outlined below.
Note: The installation of this software version will have no effect on the firmware version – the firmware version will remain at the existing level.
Issue addressed in Version 1.11
Intermittent occurrences of screen freezes has been addressed by version 1.11
Downloading the firmware file:
Download the firmware file on a PC or laptop and unzip the new software and copy to your USB disk. Make sure that no other files are present on the memory stick.
On the USB, the “Neon” folder should be present. If there is not a “Neon” folder, you should create the folder. The update files (which includes 4 files and 1 folder) should be in the Neon folder.
With the power turned off, open the USB cover on the back panel of the Neon.
These products may be covered by one or more Limited Use Label Licenses. By use of these products, you accept the terms and conditions of all applicable Limited Use Label Licenses.
The trademarks mentioned herein are the property of Life Technologies Corporation and/or its affiliate(s) or their respective owners.
LIFE TECHNOLOGIES CORPORATION AND/OR ITS AFFILIATE(S) DISCLAIM ALL WARRANTIES WITH RESPECT TO THIS DOCUMENT, EXPRESSED OR IMPLIED, INCLUDING BUT NOT LIMITED TO THOSE OF MERCHANTABILITY, FITNESS FOR A PARTICULAR PURPOSE, OR NON-INFRINGEMENT. TO THE EXTENT ALLOWED BY LAW, IN NO EVENT SHALL LIFE TECHNOLOGIES AND/OR ITS AFFILIATE(S) BE LIABLE, WHETHER IN CONTRACT, TORT, WARRANTY, OR UNDER ANY STATUTE OR ON ANY OTHER BASIS FOR SPECIAL, INCIDENTAL, INDIRECT, PUNITIVE, MULTIPLE OR CONSEQUENTIAL DAMAGES IN CONNECTION WITH OR ARISING FROM THIS DOCUMENT, INCLUDING BUT NOT LIMITED TO THE USE THEREOF.
For Research Use Only. Not for use in diagnostic procedures.